Identifying potential use errors
When working with usability engineering and interface design, a very effective and systematic way to help you identify potential use errors is to use the ‘PCA analysis’. The PCA analysis breaks down each use step into three components: Perception, Cognition, and Action, hence the name PCA.
Breaking each step of the users interaction with the device down to these three components, will help you understand what each task requires from the user.
A use error or difficulty is likely to occur if the user is unable to meet one of these task requirements related to perception, cognition and action. In this way, the PCA analysis, can be a very usefull tool to help you identify use errors related to hazardous situations.
User-device interaction model
- hazardous situations caused by a response of the medical device and
- hazardous situations caused by a user action or lack of action based on information obtained from the medical device.
PCA analysis provides valuable input to the user interface design
We have introduced the PCA analysis and the user-device interaction model. Let’s consider a simple example of how to apply the PCA analysis in order to both (1) identify potential use errors and (2) identify design input requirements:
Power on device
Perception – detect ON/OFF button
Cognition – understand that pressing the ON/OFF button powers on the device
Action – apply pressure to depress the ON/OFF button
The user cannot detect the button
The user does not understand that pressing the ON/OFF button powers on the device
The user is unable to apply enough pressure to depress the button
Device is not powered on
Design input requirement
ON/OFF button shall be backlit
ON/OFF button shall use standard icon according to IEC 60417-5009
The force required to activate the ON/OFF button shall be between 3.1 – 22.2 N (0.7 – 5 pounds)
Want to know more?
Insights you might also be interested in:
Why you should apply virtual DoE to the design process
Early product insights and fewer physical tests are some of the benefits of using Design of Experiments in medical device development.